Corventis 483 Observations Note Design Control Issues
Corventis’ design validation didn’t ensure its Nuvant mobile cardiac telemetry system conformed to defined user and patient needs and intended
uses, according to a Form 483.
300 N. Washington St., Suite 200, Falls Church, VA 22046, USA.
Phone (703) 538-7600 - Fax (703) 538-7676 - Toll free (888) 838-5578.
Copyright by FDAnews. All rights reserved. Do not duplicate or redistribute in any form.