Home > Newsletters > The QMN Weekly Bulletin > Software Glitch Triggers Class I Recall of Hospira Symbiq Infusion Pumps
The QMN Weekly Bulletin
Nov. 9, 2012 | Vol. 4 No. 45
Software Glitch Triggers Class I Recall of Hospira Symbiq Infusion Pumps
Hospira is voluntarily recalling an infusion pump due to reports the device’s touch screen control panel may become unresponsive, according to a recent FDA bulletin.
The nationwide Class I recall, announced Oct. 29, involves one- and two-channel versions of the Hospira Symbiq Infusion System from all distribution dates.
Lake Forest, Ill.-based Hospira first informed customers of the recall in an Aug. 29 letter. While not requesting that the units be returned at the time, the letter advised operators to consult the instruction manual and double-check the dosage on the touchscreen before beginning therapy.
Simple manufacturing mishaps can lead to recalls costing devicemakers thousands of dollars. Regulatory agencies around the world could change what manufactures must do when a recall is performed, but FDAnews’ Devices & Diagnostics Letterwill keep you informed on any change or guidance these agencies might offer.
For more than 35 years, Device & Diagnostics Letter has been a must-read for leaders in the device and diagnostics industries needing to stay in compliance and avoid costly design and production mistakes. Subscribe today!
ePublishing :: CMS, Hosting & Web Development | © Copyright by FDAnewsAll rights reserved. Do not duplicate or redistribute in any form.