Simplifying Global Compliance
Johnson Matthey Pharma Services - January 15, 2010
The firm failed to perform Performance Qualification (PQ) for their High Pressure Liquid Chromatograph (HPLC) systems. For all Agilent 1100's Series HPLC systems; no performance qualification is performed on the detector, pump, oven and auto injector. In addition, firm failed to monitor the lamp hours and the energy of their lamps after hours of operation as recommended by the manufacturer.
Copyright ©2019. All Rights Reserved.
Design, CMS, Hosting & Web Development :: ePublishing